Autoclave Maintenance And Use
Autoclave | Autoclave price | high-pressure steam autoclaves | autoclaves Price | Autoclave use | Sanyo Autoclave | Sanyo Autoclave | Autoclave Quote | Automatic autoclaving bacteria pot | portable autoclave | Sterilizer medical
1 , routine maintenance
1.1 sterilizer appearance and sterilizing chamber to keep clean and dry.
1.2 probes, water level should be cleaned regularly .
1.3 doorframe , apron without damage , steam inlet must not be blocked . Preferably every day after use in strips coated talc. To extend the tape life.
1.4 The interlock device to be flexible , reliable, open freely .
1.5 traps cleaned once a month , in order to facilitate discharge air to maintain the temperature .
2, the maintenance cycle
After one year , every year to ask a qualified inspection department to do a comprehensive and systematic examination . Including : Simplified , doors , piping systems , electrical systems. Safety valves, thermometers , pressure gauges should regularly check to ensure equipment safety and proper use .
3 , parking
Such as long-term parking , the device should be placed in ventilated, dry place, not the rain , when necessary, a shelter . Be drained water in the steam generator , and the door is open , sterilizing chamber to be kept clean and dry .Check more details on http://www.autoclavesale.com/ .
4 Precautions for daily use
4.1 sterilizer should be trained and qualified personnel to operate , the entire sterilization process should be supervised .
4.2 can not fully rely on automatic water level protection , you should always pay attention to the water level , so as not to burn out the heating pipes .
4.3 When artificial water . Should cut off the power , open the pressure relief vent valve 3 , again entered the water valve . Do not open when there is pressure on the mezzanine inlet valve , add water when the vent valve should be left open .
4.4 When the sterilization chamber pressure , the interlock lever can not be raised, can not be forced to open the door .
4.5 pairs of liquid sterilization should slow release steam . Until the liquid temperature drops below 700C in order to open the door. Prohibit door immediately after sterilization .
4.6 sterilization process if there is a power failure or other causes than sterilization temperature, sterilization temperature should be reached again when the temperature begins timing .
4.7 When there is pressure on the mezzanine weekly pull valve handle several times to ensure that the safety valve is working properly.
4.8 Daily classes will sterilizer glove plate away, with a dry sterile cloth set and maintain sterilization chamber dry.
4.9 Class day after the steam generator water with pressure discharge end, in order to reduce the scale .
4.10 Chang Lu equipment operation should not be required to build when Chang Lu protective frame , to prevent rain .
4.11 Continuous use of time should be less than 8 hours / day .